Valentine (1996), which apparently overcomes some of the disadvantages of the system described already. The sensor utilises what was described as a selective narrow band in the near infra-red, which allowed good discrimination between yeast cells, other particulates and gas bubbles. The linear operating range extended to 2 x 109 cells ml 1 and therefore would be suitable for measurement of yeast concentrations in pitching slurries. This would allow the setting up of an automatic pitching system in which the concentrated slurry could be added quickly, as in conventional pitching, and overcome the problem of having to dose in yeast throughout wort collection. It would still be necessary to correct for viability by separate analysis.
More recently, the application to the enumeration of suspended yeast particles of another optical system has been described (Holmes & Teass, 1999; Holmes, 2000). This system was applied to the control of yeast pitching rate in an automatic system. In comparison to an older manual system based on laboratory analysis, improvements in fermentation consistency were reported.
126.96.36.199 Use of radiofrequency permittivity. A biomass sensor has been developed which detects cells by virtue of their dielectric properties (Harris & Kell, 1986; Harris et al., 1987). Suspensions of cells possess several recognisable frequency-dependent dielectric dispersions (Schwan, 1957). One of these, is called the (3-dispersion. It is caused by a combination of build up of charge at the surface of the relatively nonconducting plasma membranes and partially restricted lateral motions of the charged lipids and protein components within the membrane. These are generated in response to an applied field (Woodward & Kell, 1990, 1991; Ferris et al., 1990). When the applied signal is of the frequency of radiowaves, the value of the (3-dispersion of a suspension of yeast cells is proportional to the volume fraction of biomass bound by a biological membrane. Since yeast cells of a given strain and in the same physiological
Head amplifier electronics module
Screw fitting retaining nut
Inert resin probe c. 12 x 2.2 cm electronics module
Inert resin probe c. 12 x 2.2 cm
Body of in-line fitting
Was this article helpful?
Discover How To Become Your Own Brew Master, With Brew Your Own Beer. It takes more than a recipe to make a great beer. Just using the right ingredients doesn't mean your beer will taste like it was meant to. Most of the time it’s the way a beer is made and served that makes it either an exceptional beer or one that gets dumped into the nearest flower pot.